Motion monitor

ABSTRACT

A system for monitoring the respiratory activity of a subject, which comprises one or more signal generating elements being kinematic or light emitting elements, applied to the thorax of a subject, for generating signals that are indicative of movement of the thorax of the subject; a receiver for receiving the generated signals during breathing motion of the subject; and one or more computing devices in data communication with the receiver, for analyzing the breathing motion. The more computing device is operable to calculate, in response to the received generated signals, the magnitude of a maximum displacement in 3D space of the one or more signal generating elements during a cycle of the breathing motion; and to calculate the magnitude of a current displacement in 3D space of the one or more signal generating elements during the breathing motion with respect to a reference tidal volume associated with the maximum displacement in 3D space.

This application is a continuation-in-part of PCT/IL2014/050916 filed on Oct. 22, 2014, which claims priority from U.S. 62/020,649 filed on Jul. 3, 2014 and U.S. 61/895,000 filed on Oct. 24, 2013.

FIELD OF THE INVENTION

The present invention relates to monitoring apparatus. More particularly, the invention relates to image-based monitoring apparatus for monitoring a variety of activities, such as human breathing.

BACKGROUND OF THE INVENTION

Respiration rate is a vital sign that is monitored either intermittently or continuously in a variety of situations, including but not limited to during and after surgical intervention. Continuous monitoring apparatus can be placed in contact with the body (abdominal belts or air flow on trachea or mask) or non-contact (e.g. pressure sensors under mattress, or a Doppler sensor). Continuous monitoring of respiration beyond the respiratory rate (such as breathing patterns, breathing depth, minute ventilation and apnea) is achieved nowadays using contact and invasive sensors.

SUMMARY OF THE INVENTION

The present invention relates to a video system for non-contact monitoring of the respiration properties through various measures such as respiratory rate, respiratory effort, tidal volumes and respiratory patterns. These properties are obtained under a wide range of ambient conditions such as in the presence of subject motion, subject out of bed, various ambient lighting conditions, obstructions, etc. In addition reliable early warnings can be issued based on such respiratory properties and based on analysis of the video and possibly additional signals over time. The term “subject”, as used herein, is meant to indicate an individual the respiratory activity of whom is being tracked, whether a patient in a health facility, or a person being tested for any reason or purpose.

The system is applicable to various settings such as monitoring subjects who are undergoing sedative or pain killing treatment that can depress respiration, monitoring deterioration in the critically ill, monitoring infants to protect against SIDS and diagnostic tools for sleep testing such as for obstructive sleep apnea. In such cases, the monitor of the invention can be used to track other movements in addition to respiratory movements, e.g., leg movements and eye movements, as well as for quantifying awakenings and sedation level. These can be important for sleep staging analysis.

Another use of the invention is for the diagnosis of conditions using intermittent monitoring or comparison of measurements taken at different times. One example of this is comparison of breathing movement patterns of the chest and abdomen before and after surgery in order to diagnose diaphragm paralysis. Also, change in respiration patterns at rest over time can be tracked with the system of the invention.

The settings in which the system may be used are versatile such as hospitals, clinics, home, battlefield and even outdoor environments. Often for some of these conditions, respiratory rate is a late indicator of respiratory distress, and other measures such as the depth of breath and pulmonary motion patterns are earlier indicators of deterioration. There are situations in which the respiratory rate is not the differentiator but rather the overall pattern of body movement.

BRIEF DESCRIPTION OF THE DRAWINGS

In the drawings:

FIG. 1 is a flow chart of a monitoring stage, according to one embodiment of the invention;

FIG. 2 is a flow chart of a detection stage, according to one embodiment of the invention;

FIG. 3 is a flow chart of a tracking stage, according to one embodiment of the invention;

FIG. 4 is a flow chart of an analysis stage, according to one embodiment of the invention;

FIG. 5 is a flow chart of a volume calibration stage, according to one embodiment of the invention;

FIG. 6 shows a marker according to one embodiment of the invention in separated position relative to a cloth, as may be part of the subject's robe;

FIG. 7 shows the marker of FIG. 6 in operating position;

FIG. 8 is a top view of the marker of FIG. 7;

FIG. 9 is a cross-section taken along the AA line of FIG. 8;

FIG. 10 is an exploded view of the disposable part of a marker according to one embodiment of the invention;

FIG. 11 is a breathing waveform showing a single marker's displacement over time;

FIG. 12 is a schematic illustration of a system for monitoring the breathing patterns of a subject using inertial sensors; and

FIG. 13 is a method for determining the respiratory activity of a subject according to one embodiment of the invention.

DETAILED DESCRIPTION OF THE INVENTION

The invention will be described below with reference to illustrative embodiments, which are not meant to limit its scope but are only meant to assist the skilled person to understand the invention and its uses. While in some cases reference may be made to a single subject or to a single marker, or to a specific number of markers, it should be understood that this is done for the sake of simplicity but the invention is not limited to the monitoring of a single subject and more than one subject can be monitored simultaneously by a single system of the invention. Similarly, different numbers of markers can be used, as will be further discussed below, which can be positioned at different locations, depending on the result that it is desired to obtain.

The following description relates to a system for monitoring the respiratory activity of a subject in a static position such as in bed, but it will be appreciated that it may also be used to monitor the respiratory activity of a moving subject, particularly one that is undergoing physical activity, such as riding a bicycle.

Exemplary System Overview

An illustrative system according to one embodiment of the invention consists of the following components:

-   -   1. One or more markers on the subject comprising signal         generating elements, power supply elements and circuitry for         operating the signal generating elements, either directly         contacting the subject or on the subject's clothing or covers         (i.e., on the sheets and/or blankets).     -   2. A receiver for receiving the generated signals, for example         an imaging device, which can be for example a CMOS video camera,         a 3D camera, a thermal imager, a light field camera or a depth         camera. The imaging device may have an associated illumination         source such as IR LEDs in some embodiments. The receiver may         also be a wireless receiver that receives transmitted signals         generated by elements, such as an accelerometer or a gyroscope.     -   3. A computing device that is connected either wirelessly or         wired to the receiver. The computing device may be remote and         the connection to it may include an internet connection. As an         example, some image analysis may take place on a wired connected         device such as a computer board wired to the receiver. From         there, a Bluetooth connection (or a connection of functionally         similar technology) may connect to a gateway device that uploads         the data to a cloud on the internet, or delivers it to a remote         location, for further processing and storage.     -   4. An optional alarming device that issues either an audible or         other kind of warning near the subject or to a remote location         through a wired or wireless connection from the computing         device.     -   5. An optional remote application, e.g., at a central nurse         station or on a mobile device that can be used to control the         signal generating element in the subject's vicinity and receive         data from the signal generating element. For example, the remote         application can be used to stream raw video or analyzed video         from the signal generating element located near the subject,         once an alert is issued.     -   6. An optional display device that may be connected with a wired         or wireless connection to the computing device.

The system may also comprise:

-   -   a. Additional signal generating elements, such as acoustic or         other sensors that are connected to the computing device, e.g.,         a pulse oximeter.     -   b. Data storage devices, which can be located on the receiver,         on a computing device or at a remote location, for storing data         such as an electronic medical file. In addition, a database         which includes training data for learning method parameters can         be located on a computing device.     -   c. A spirometer or respirometer which measures a subject's air         flow for calibration purposes in some embodiments.         Physiological Measurements         Extracting Breathing Measures

Accepted clinical measurements such as respiratory rate, tidal volume, minute ventilation and apnea indicators can be derived from the sensing system. In order to obtain absolute estimates of some of these properties (typically for diagnostic purposes), calibration is needed between the measured signals and the respiration property. These calibration functions can be learned a priori based on subject attributes such as shape, silhouette, weight, age, sex, BMI, chest circumference, and body position and data from a validated device such as a spirometer. The calibration can also be updated using data from a phantom or a simulator. Note that calibration functions used should be taken with the subject pose taken into account.

Often for monitoring, the absolute value is not necessary but changes from baseline properties need to be quantified. In such cases, the calibration need not be carried out. In the following, methods for extracting a number of respiratory properties from the video sensed markers are outlined:

-   -   1. Respiratory Rate: The respiratory rate is extracted by first         extracting the dominant frequency from each marker separately         and then fusing these estimates together in some way. One method         to extract the dominant frequency from the marker's cleaned         (filtered, cleaned of noise and after non-breathing motion         removal) signal, is through analysis of time series windows         which include several breaths. For instance, this can be         achieved by calculating the Fourier transform over a moving         window of, e.g., ˜20 sec every 1 sec. A Hamming window         (Enochson, Loren D.; Otnes, Robert K. (1968). Programming and         Analysis for Digital Time Series Data. U.S. Dept. of Defense,         Shock and Vibration Info. Center. p. 142) is used to reduce         artifacts. The instantaneous frequency of each marker is taken         to be the frequency with the largest Fourier component         amplitude. Another way of calculating the individual respiration         rates are in the time domain as the distance between consecutive         significant peaks or troughs.

The estimate of the actual respiratory rate is taken as a weighted average of the dominant frequencies found from each valid marker. The weighting can be taken proportional to the amplitude of the marker motion and outliers can be iteratively removed if sufficiently far from the average rate computed using the current subset of markers.

-   -   2. Apnea and Hypo-apnea Events: These are events for which the         subject has stopped breathing or has a very low breathing rate         for over 10 seconds.     -   Apnea can appear with chest wall movements present as in an         obstruction and there a more careful analysis is needed. First         the amplitude of the chest movements is examined and if it is         significantly lower than recent breaths it is suspected as an         event. Furthermore, the phase difference between abdominal and         thoracic markers is followed to determine whether a significant         change 1 time may be obtained by obtaining the phase delay         between two short time segments (10 sec for example) of the         marker movements. For example, this can be achieved by taking         the Fourier Transform of each segment separately and obtaining         the phase difference between the relevant complex Fourier         component of each of the two markers. The relevant component is         the one that correspond to the extracted respiratory rate.         Another method to find the phase difference between two markers         is to average the time lag between their peaks and express it as         a fraction of the cycle time; 50%, for example, corresponds to a         180° phase lag.     -   3. Inhale time, Exhale time: For each marker displacement trace,         the time evolved between a peak and the following trough can be         used as a measure of the inhale or exhale time, depending on the         marker location.     -   4. Breathing Variability: Breathing variability can be         calculated as the entropy of the lengths of the breathing cycle.         It is often an indicator of respiratory distress.     -   5. Tidal Volume: A method for calculating tidal volume (the         amount of air displaced or inhaled per breath) via calibration         on a training set and learning a function on displacements was         described above.

Another method for calculating the tidal volume is through modeling the abdominal and thoracic cavity shape. For example the shape can be assumed to be a double cylinder. Both cylinder walls expand during inhalation and contract during exhalation. The heights of the two cylinders change with breathing due to the movement of the diaphragm. The thoracic cylinder and abdominal one have different mean radii which can be approximated from the image by using edge detection on the image to find the body edges, at one or more body positions. The known physical marker size can be used to convert these measurements to physical units. During runtime, the marker displacements and the angle of the camera are used to calculate tidal volumes through the model.

Yet another modeling technique can utilize the 3D modeling of the thorax as described hereinabove. The extracted marker motion can be superimposed on the 3D model to extract the volume change during exhalation.

-   -   6. Minute Ventilation: Integrate tidal volume over one minute.         Extracting Non-Breathing Measures

There are other measures of motion that can be found from the monitoring system that are not associated with breathing, but can give information on sleep quality and behavior that can be extracted from the marker movements. Some of these are:

-   -   1. Leg Movements: By tracking markers on legs or other         extremities the extent and number of these movements can be         calculated. Such movements are associated with the diagnosis of         restless leg syndrome.     -   2. Eye Movements: Eyes, when open can be detected from the video         imaging without the necessity of markers. In order to classify         eye movements during sleep small markers placed on or above the         eyelids can be used. These may be used for sleep staging and         also for quantifying the continuity of sleep (eye openings).     -   3. Posture stability: The number of position changes during         sleep can be tracked easily due to the markers that move in and         out of the field of view. The actual body position at each time         of sleep can be found and alarms can be set to caregiver if         person needs to have position changed (subjects with bedsores,         infants who are not to be positioned in prone positions, etc.).     -   4. Bed Exits: The number of bed exits and bed entries can be         detected and alarms setoff in real time if person does not         return to bed or falls near bed for example.     -   5. Heart Rate: The heart rate can be obtained from markers         located near to chest. This can be best done by finding peaks in         the Fourier transform of the displacement signal of each marker         in the range of, e.g., 40-140 bpm for adults.     -   6. Tremors/seizures: Continuous trackable motion in all the         markers as well as motion in a large fraction of pixels found         within the body silhouette indicate a tremor.     -   7. Eye micro-tremor: The characteristics of eye micro movements         are related to the level of sedation. The described sensor can         be used in conjunction with a marker to track these movements as         follows: A marker can be adhered to the eyelid, such as a thin         retro-reflective adhesive sticker or else IR reflective polish         can be applied to eyelids or eyelashes. An alternative is to         attach the marker to the side of the face near the eye and have         a thin flexible object connecting it to the eyelid. It is         possible to track movements by detecting and tracking the         boundaries between eyelashes and eyelids in the image. Since the         movements are of high frequency (˜80 hz) and of small amplitude         (microns). The choice of camera sensor should be made         accordingly and also the placement should be closer to narrow         the field of view. One possibility is to attach the camera to an         eyeglass frame rather than to the bed.

In order to cancel camera jitter movement and head movements which, other reference points away from the eye are also tracked such as markers on the cheek, forehead or earlobe. These signals are analyzed to detect the main frequencies of motion which can subsequently be removed from the eyelid signal. The resulting signal can be further analyzed to determine the frequency of the micro eye tremors.

The eye-tremor and breathing monitors can be configured to use the same sensor or distinct sensors.

-   -   8. Head Movements: Often there are head movements arising from         an obstruction and manifesting itself as snoring and increase         respiratory effort.         Respiration Quality Measures and Warnings

Respiration is monitored in order for an early warning alert to be issued if a problem is about to occur. The system produces an online early warning alert based on personalized per subject data and based on recent continuous measurements.

The ubiquitous problem with alert systems is that they are often unreliable and therefore lead to alarm fatigue, in which alarms are often ignored. We describe a system which provides a more reliable early warning, is more immune to artifacts and classifies the overall respiratory information sensed. In addition, a quality measure is provided, which quantifies the reliability of the current measurement. The early warning system is based on a multi-dimensional analysis of the respiratory waveforms and personalized to the current attributes of the subject being monitored. This should be contrasted to typical monitoring systems that track a single feature, for example the respiratory rate. In such systems, an alarm will typically be set off whenever the derived respiration rate declines or exceeds fixed preset minimal and maximal thresholds. The approach that introduce here is different in that it is adaptive to the subject but reliable in that it is based on a large amount of data. First, features from the subject are used (age, sex, thoracic size, etc.) and then various feature are extracted from the recent measurements. These can be the outputs from the video imaging system described above or from other devices or from a combination of several devices. For example, the raw signals may include those from an end tidal CO₂ capnograph, a pulse-oximeter and the video system described above with a single marker. The measurements can be impedance measurement made through leads attached to a subject. The identity of the features, mathematical quantities derived from the measurements time series, are determined in the training stage described below. A classifier is trained in this feature space during the training stage which is performed either in advance or online. The classifier can be a two-class one that differentiates between normal subject behavior and abnormal behavior. Using the classifier, a score is assigned continuously based upon the measurement data, with 1 indicating typical baseline behavior while a lower score, e.g., 0.5, represents the edge of normal behavior below which outlier behavior is observed. In addition to this “functional score” that is calculated based on training data, a quality score is calculated denoting how reliable the basic measurements are in cases where this is possible.

Signal Features

The features are calculated from the various signals on various timescales and typically depend on the physiological source of the signal. For example, the respiratory rate calculated from a 20 second time series signal of several of the video markers described above can be one of the features. Other features can quantify the trend of physiological quantities, such as the derivative (trend) of the respiratory rate over consecutive overlapping 20 second intervals. Similarly, average amplitudes of the respiratory signal peaks and troughs over 20 second intervals and their derivative over time can be used. The initial set of features can be reduced further in the training stage to reduce over-fitting of the classifier using standard methods.

Training

A classifier is trained using training data, which consists of training vectors that consist of the set of features and the label of one of the classes to be classified. The label may be “normal”, “coughing”, “sensor disconnected”, “risky”, and so forth. If sufficient data exists for more than one-class of behavior, a multi-class classifier map be trained. Minority classes with few data points can be grouped to a larger class labeled “artifacts” for example. In the case where labeled data does not exist or predominantly belongs to a single class, a single class classifier can be trained. Frameworks for learning such classifiers exist, for example the SVM one-class classifier (B. Scholkopf et. al., Estimating the support of a high-dimensional distribution. Neural Computation, 13(7), 2001).

Standard feature selection methods can be used to reduce the dimensionality of the classifier problem, such as PCA, and methods that rank the features according to their discrimination ability. The forgoing applies in general to training a classifier based on a fixed training set that is collected a priori to monitoring. However, if the data is collected a priori, it may not be relevant to the baseline characteristics of the subject in question, thus leading to an inaccurate classifier. An alternative would be to train the classifier based on the initial period of monitoring the subject, e.g., the first few minutes when the medical staff is in the vicinity. However, this greatly limits the amount of training data that can be used and thus can severely limit the generalizability of the classifier due to over-fitting. Again the classifier results may not be reliable. Our proposed methods involve forming training sets which are relevant to the subject in question and enable the incorporation of a large amount of data. Several variants which can be used alone or integrated together of forming a training set are outlined below:

-   -   1. Augmenting training sets online: The data from a monitored         subject can be added to the database by adding it as time         segments of e.g., 10 minutes with all its features. In addition,         the periods of respiratory depression can be labeled according         to preset conditions such as respiratory rate <8 or minute         volume <3 liter. Using such a criterion, all the data can be         labeled according to how long they precede a respiratory event.         This data can be used in future training sets of the classifier         since it can be labeled as “normal” and “risky” depending on how         long before a respiratory event they were extracted.     -   2. Subject training set selection “on the fly”: One method is to         train a classifier based on data from a subset of the total         available subjects. This subset is chosen so that it is         “closest” to the subject in question. One method this can be         done is by carrying out a clustering procedure (e.g., mean-shift         clustering) in the feature space composed of the data from all         the subjects, including the initial data of the subject in         question. The subjects, whose data falls mainly in the same         cluster as those of the subject are those used as the training         set. In order to improve stability yet further the clustering         can be carried out many times with different parameter values;         the number of times each subject falls in the same cluster as         the current subject is recorded. Subjects that fall in a         majority of the experiments in the same cluster as the subject         in question are used for the classifiers training set.     -   3. Training set selection “a priori”: The data from all subjects         can be a priori clustered as in method 1 to obtain subsets of         subjects with similar physiological signal features. More than         one clustering experiment can be used to produce additional         subsets that may intersect previous clusters. Each such cluster         is a candidate training set and the one that is chosen is the         one who is closest to the current subject. This subset can be         chosen according to the distance of the cluster's center of mass         to the current subject's features mean value. In addition,         differences in the principal component directions in feature         space can be incorporated into the distance matrix between the         two distributions.

Another method of finding the distance between the current subject and the subsets is to determine the overlap volume between each cluster distribution and the current subject distribution. Each such distribution can be approximated by an ellipsoid using the covariance matrix and the center of mass in feature space. Additional criteria than can be taken into account in calculating the closeness are subject attributes such as age, medical status, sex and base vital signs.

-   -   4. Transforming training set: Another option of utilizing the         original training set for the new subject is to scale the         individual subjects to normalized coordinates. For example, a         feature value f_(i) can be transformed as follows:

$\begin{matrix} {f_{i}^{\prime} = \frac{f_{i} - \mu_{i}^{(p)}}{\sigma_{i}^{(p)}}} & (4) \end{matrix}$

-   -    where (μ_(i) ^((p))), (σ_(i) ^((p))) are the mean and standard         deviations of the feature i for subject p. After transformation         the classifier can be learned on the transformed data set.     -   5. Selecting time segments for training set: Using the labeling         of time segments as described in #1 above the relevant time         segments from the subjects who were chosen to be included in         training set are used.

The output of the training stage are mathematical formulae whose input are feature vectors and whose output is a class probability—the probability that the feature belongs to a specific class of the classifier. In addition, regions of the feature space that are associated with non-normal respiration patterns are assigned informative clinical descriptions: such as “Apnea”, “Shallow Breathing”, etc. These regions and abnormalities are assigned a priori based on pre-assigned rules (based on expert rules).

Runtime

In runtime, an initial class is assigned continuously based on the class with the highest score of the learned functions. In cases where a global normalization is applied to the raw features, the normalization factor is updated during runtime on the previous data of the subject in question. There are many adjustments that can be made in order to improve reliability of the monitoring such as:

-   -   1. In cases where the class association is not clear-cut an         “undecided” tag can be associated with this time.     -   2. A quality measure is associated continuously in time. For the         video sensor, whenever non-respiratory body motion is         identified, the quality score is reduced from 1. The amount that         it is reduced depends on the prediction error of the motion, the         larger the error the more it is reduced (the minimal quality         measure is 0). Also if periods of coughing or talking are         identified using other sensors (for example acoustic sensors),         the quality measure is also reduced. Any indication of artifacts         in one of the sensor signals should be used to reduce the         quality measure of the other sensor signals.     -   3. Due to missing measurements (subject out of bed), it may not         be possible to calculate the classification at specific times.         The class assigned at these times can be tagged as “unknown”.

When the classier output is not “normal”, the suspected anomaly can be output based on the feature values and predetermined rules. Examples of such rules are:

-   -   1) High respiration rate (28 breaths/sec) with low variability         (0.2);     -   2) Hypoventilation: Breathing effort is low (Amplitude 50% of         normal) and respiration rate is low (5 breaths/min);     -   3) Obstruction: Paradoxical breathing.         Output: Alarms and Communication

Typically, a class output that is neither “normal” nor “unknown” can trigger an alarm or warning to be issued. In order to increase reliability, a delay time (e.g., 10 seconds) may be applied before an alarm is issued. If during this delay time, the measurements return to the “normal” class, the triggered alarm will not be issued. Alarms can be audible with sound associated with severity and type of breathing pattern. They can be communicated to a variety of output devices either by a wired or wireless connection, such as to a bedside monitor, a handheld monitor, a mobile unit or a central nursing station.

Subject Monitoring Process

The subject monitoring process is carried out according to one embodiment of the invention when a subject 5 in supine position is lying on a bed or on any other substantially horizontal support surface 7, as illustrated in FIG. 12.

A rigid marker 51 provided with two inertial sensors 56 and 57, a single inertial sensor or any other suitable number thereof, such as a three-axis accelerometer and a gyroscope, or a combination thereof, is used to calculate the respiratory activity of subject 5. Marker 51 is applied to the thorax 13 of subject 5 by engaging means 16, such as an adhesive connection applied directly to a bodily portion or an interfacing element such as a fabric in engagement with the thorax, to define a desired positional relationship to a chosen location on the subject's thorax. Each inertial sensor may be equipped with its own power supply element 22 and operating circuitry 24, or alternatively marker 51 has common power supply elements 22 and operating circuitry 24 for all inertial sensors provided therewith.

During breathing motions, thorax 13 is displaced cyclically in 3D space. Marker 51 is also displaced by a distance of d1 in 3D space in response to the thorax displacement. Inertial sensors 56 and 57, as they are displaced, sense the force applied by thorax 13 during a breathing motion along a local inertial frame defined by their rectilinear structure, and in turn output a signal S that is indicative of the detected force or of the acceleration related to the detected force. Output signal S is generally a wireless signal, but may also be transmitted by a wired connection. Receiver 33, which is located in the proximity of marker 51 when the generated signal is a wireless signal but which also may be remotely separated from the marker, receives the generated signals S and transmits them to a computing device 37, which processes the generated signals S to calculate a sensor displacement. Alternatively, the receiver 33 may be wired and located in close proximity to the computing device. Both of them may be situated on or near the patient's bed.

The motion of the rigid marker 51 in a single exhalation or inhalation movement may be approximated as a 3D translation, and therefore it may be assumed that the local accelerometer axes are also fixed during an exhalation or inhalation movement. However, the gravitational field adds a fixed acceleration value to the output signals S, which may lead to incorrect conclusions regarding the respiratory activity. A high pass filter provided with computing device 37 may be used to filter output signal S and to thereby remove the fixed gravitation-based acceleration value.

FIG. 13 illustrates a method for calculating a subject's current or average breathing movement and/or current tidal volume using the signals generated by the inertial sensors.

After the computing device receives the signals generated by the inertial sensors in step 62, each signal is filtered in step 64 twice to remove two sources of error: the fixed gravitation-based acceleration value, and also a determined rotation component associated with the subject's motion, imost prominently non-breathing motion. Non-breathing motion may be identified according to the presence of significant rotations in the signal related data; a possible criterion is that the cumulative angle of rotation over a predetermined period of time, e.g., the previous 10 seconds, is greater than a predetermined threshold. A Kalman filter, or any other filter providing high sampling rate measurements of the output acceleration values, is suitable. After filtering, the thorax-caused motion can be approximated as being translational.

Each filtered output signal generated between the two extrema that represent acceleration over time related to exhalation and inhalation movements, respectively, is integrated twice in step 66 to first calculate a change in velocity and then a change in breathing displacement as a function of the integral. In order to reduce integration error, the integration is carried out between peaks and troughs of single breaths by enforcing regularization constraints such as zero velocity at peaks and troughs of the respiratory signal. Then, in step 68, the magnitude of a current breathing displacement or a current average tidal volume during a cycle of breathing motion is determined, and in step 70, is compared with the stored calibration function, in order to analyze the respiratory activity.

Another embodiment of the monitoring process is carried out by imaging, and includes four major steps and a calibration step (the accelerometer will need a calibration step as well in order to output tidal volume. This can be done using the described calibration with camera. If no camera is available one can consider doing a calibration involving only the non-camera measurements (age, weight, etc.) and accelerometers. More accelerometers can be used for calibration to obtain body shape), which will be described below with reference to FIGS. 1-5.

Sensing and Data Acquisition System

Video Sensing:

The imaging system of this particular embodiment of the invention consists of the following components:

-   -   1. Sensors and Illumination: One or more video cameras are used         to record a scene of the subject. One example is a CMOS camera         with night-vision capabilities, i.e. one in which a filter         exists to filter out the visible light and sense the near IR         optical wavelengths (NIR: 0.75-1.4 μm). There may also be         ambient infrared LED illumination to increase ambient light if         it is low. The ambient light source may be on the camera or in a         chosen spatial arrangement in the subject's vicinity. Another         example for the camera sensor is a thermal camera which is         sensitive to mid-wavelength (3-8 μm) or long-wavelength (8-14         μm) infrared radiation. It too can have a light source such as a         halogen lamp source which is used in Thermal         Quasi-Reflectography. The sensor may also be a 3D depth camera         such as the Carmine 1.09 by Primesense Corporation.     -   2. Markers: Markers are applied either directly to the subject's         body or integrated to a covering of the subject such as his         clothing, his blanket, an elastic strap, or a bandage for         example. The markers can be for example patches made from         retro-reflective material, geometric patterns on the blanket or         nightgown, or low voltage LED lights embedded or attached to         clothing or patches. Each LED light can be either fixed or can         flash each with its own known temporal pattern. Also, their         light emitting wavelength and illumination may be specific for         each LED. In order to enhance the outward facing light of these         active markers, they can be embedded in retro reflective cones         or other shaped contraptions made of reflective materials. Also         the illumination angle may be narrow by using a lens. One         specific marker, which is an object of the present invention and         which can be used in a system such as this illustrative system,         will be described in greater details later in this description.

Several markers may be incorporated together such as into an adhesive linear strip, or a 3D shape such as a dome shape consisting of scattered markers on its curved surface. Another possibility is to design the blanket that covers the subject to include “windows” that are transparent to IR light at the locations of the markers. The blanket can be fastened to the gown or marker patches to keep the markers visible through the windows.

In one embodiment of the invention the marker unit consists of two distinct physical parts that can be firmly connected to each other but can also be easily released from each other. One part may be positioned close to the body and may therefore be for single use (cannot be disinfected) while the other part (can be disinfected and therefore reusable) attaches to the disposable part but has no direct attachment to the subject. The attachment between the two parts can be direct through fasteners, e.g., similar to ECG fasteners. Another alternative is that the two parts be connected through clothing or a blanket. For example, in one embodiment of the invention the disposable part is a patch that adheres to the body, while the reusable part consists of an active marker that is positioned on clothing and attaches to the disposable part through one or more fasteners. In the case of an active marker, the power unit may either be incorporated within the disposable patch or within the reusable marker.

The active markers can emit steady signals or coded signals. For example if the markers emit light, the intensity can be modulated according to a frequency that is predetermined and distinct for each marker location. In this way, each marker can be easily identified during operation. Furthermore, light patterns which can save on battery life can be employed for long-term monitoring.

-   -   3. Controller and Mount: The camera can be hung overlooking the         subject on the wall, ceiling, a stand or part of the bed or any         other fixture in the vicinity. A controller may be provided,         which can be used to pan, rotate and zoom the camera's field of         view freely in 3D either manually or automatically based on the         video content. In addition to the camera positioning, it can be         possible to adjust other camera features automatically based on         the video content. Examples of such features are: focal         distance, illumination level, white balance and optical filters.         Providing such controlling elements is well known in the art and         is not therefore described herein, for the sake of brevity.

Many different embodiments of the invention can be provided, comprising different elements, such as sensors, software for calibration or other purposes, etc. Some such elements are described below, it being understood that invention is not limited to embodiments using such elements, and that many other additional functional elements can be added, which are not described in detail hereinafter since they will be understood by the skilled person.

Additional Sensors

The camera visual sensor may also be equipped with a depth sensor providing the distance of each monitored pixel from the camera. Other cameras may provide the lightfield at each pixel, which in addition to the scalar RGB values provides the direction of the light incident on the sensor's pixels (see for example the lytro at http://www.lytro.com).

Video Detection and Tracking

The following relates to the detection and tracking of the markers using 2D video images (FIGS. 2 and 3). Later on a description is provided how the depth information can be obtained from standard 2D images or from a 3D camera or stereo camera For the first video frame or any frame in which the markers from the relatively recent frames were not all tracked, an attempt to detect the markers is carried out. Detection on the intensity image (or sequence of images) can proceed in the following steps:

-   -   1. Pixels with high intensity are marked. This is done using a         fixed threshold or one that adapts to the histogram of the         entire image. For example the gray level of the 90^(th)         percentile of the image histogram is found; only pixels which         surpass this value by a set number of standard deviations are         marked. The standard deviation is calculated from the image         histogram.     -   2. The marked pixels are clustered to connected components using         chain clustering for example. Two pixels belong to the same         cluster if there is a path of marked pixels that connect them,         each within distance d of each other.     -   3. The various clusters are the candidate markers. They are now         filtered according to a priori known features of the markers         such as their size, eccentricity, bounding box size, temporal         frequency and gray levels. For circular markers, principal         components values of the candidate clusters are useful features.         Position of the candidate clusters relative to the bed or body         silhouette can also be used to filter out spurious clusters.

In the case where the markers are “flashing” and do not appear in each frame, the above analysis is carried out for a series of frames where the extent and identity of the frames in the series depend on the flashing frequencies. The final candidate markers for step 3 above consist of the unified set of markers obtained from the frame series. In the cases where the marker produces a variable shape and dispersed light pattern the clustering is carried out differently. This can occur for example in the presence of a blanket which may scatter the light reflected or emanating from the markers, dependent on the folds in the blanket and the relative configuration of the blanket relative to the markers. In these cases, chain clustering with a single threshold on the intensity image as described above may not be sufficient to identify a single marker consistently. A marker may be composed of several disjoint clusters which are the “seeds” of its more extended light pattern. In order to capture the marker signature, the seeds may be expanded by region growing techniques to unite the seeds to a single signature. This signature is saved as a set of pixels with their gray levels. In addition the presence of a specific temporal behavior for a particular marker is used to associate disjoint pixels to a common signature.

Once the clusters in a frame are detected, they are tracked over the following frames (as described below). If the tracking was not successful for all the detected markers, the detection is reinitialized on the new current frame. Successful tracking means that the positions of all the detected markers were determined in each frame, that the frame to frame displacement of each marker does not surpass a predefined threshold and that this displacement is consistent across the marker pixels. The threshold is set using the maximal expected speed of the breathing motion. It can be later adjusted per patient during the baseline setting per patient described below.

In case the tracking was successful for a short time, say 3 seconds, the tracking is continued in time on the “valid” detected clusters. A valid cluster is one whose motion is consistent in terms of frequency of motion with the major detected markers (those in the trunk of the body). This can be analyzed by calculating the largest component of the Fourier Transform of the cluster displacements over say 10 seconds and removing clusters whose frequency is far from the median frequency of the chest & abdominal markers

Once it is verified that markers in the field of view have been detected as described, one proceeds with tracking these markers in the coming frames. From time to time detection is carried out again to detect markers that were momentarily absent or had a low signal in the initial detection.

The tracking of the markers can proceed by calculating the optical flow patterns in order to identify the local motion within the image. The Lucas Kanade method can be utilized (B. D. Lucas and T. Kanade, An iterative image registration technique with an application to stereo vision. Proceedings of Imaging Understanding Workshop, 121-130, 1981).

The tracking is carried out relative to a reference frame in order to avoid accumulating frame-to-frame errors. The reference frame is updated from time to time due to the fact that the scene is changing over time. One possibility is to update the reference frame when one or more of a specific set of criteria are met. Examples of these criteria are:

-   -   Non-breathing movements are identified;     -   Illumination conditions of the scene have changed significantly         (total intensity change of marker cluster relative to same         marker in reference frame);     -   Every preset time period (e.g., 30 seconds)

In order to improve the sensitivity of marker detection, the tracking can be used as a filter as follows: the single frame detection method may be set to produce an abundance of objects some of which are not actual markers on the subject's body. Once they are tracked, criteria for filtering them can be used based on characteristics of their movement. For example, markers that remain static for a long time even when subject makes some position change can be filtered. Furthermore, if one is concerned with tracking only breathing motion, non-static markers with a different frequency of motion (or uncorrelated motion) than the identified breathing markers are filtered.

Movement of non-marker features can also be tracked. For example, using IR illumination, the subject eyes are visible and can be tracked in a similar manner as an artificial marker. In this way, subject's awakenings can be quantified through measurement and recording of the times the subject's eyes are open.

From Video to Respiration

For each marker that is detected or tracked in a frame, its center of mass image coordinates are calculated by averaging the locations of the pixels in its cluster. The raw data of positions is processed further in order to reduce noise due to the measurement systems and subject motion. Furthermore the resulting motion can be classified as breathing or non-breathing motion.

Noise Reduction

Each individual marker time series (2-dimensional x and y) is processed on-line using noise reduction filters, such as those using a measurement window around each point to process the data using a mathematical operation such as averaging, weighted averaging, Kalman filtering or the median operation. Bandpass filters damping high frequency noise can also be applied.

Dictionary learning methods can also be used to reduce noise in the time series either prior to the elimination of non-respiratory motion or succeeding it. These methods are useful to reconstruct segments of the tracking signals using a small number of short time segments chosen from a dictionary which represents an over-determined basis set for breathing (see for example: Ron Rubinstein, Alfred M. Bruckstein, and Michael Elad, Proceedings of the IEEE 98(6) 2010). The dictionary is learned from time signals taken from a training set. This training set may be collected either a priori on a set of healthy subjects or from the baseline measurements on the subject in question. Additional methods of sparse reconstruction algorithms can be used to reduce noise in the original signal.

Another possibility of isolating breathing motion from other motions, such as body movement, heartbeat or body tremors, is to first learn the principal image direction of the breathing vector for each marker when the subject is at rest. All motion in the perpendicular direction is regarded as “noise” and removed. Other physiological information can be extracted from this removed “noisy component” such as the frequency of cardiac oscillations (heart rate) and tremor frequencies. Heart beats can be identified using band pass filters associated with frequencies learned based on subjects data or from an additional hear rate sensor (for example the pulse-oximeter).

In the case that the camera is mounted on a non-stationary devices such as a pole, a lamp arm or another moveable accessory, additional noise may enter the video. Such noise can be tracked by following the movement of a stationary object such as a wall or floor or a marker on a wall. Another way the movement can be tracked is by using a 3D actuator connected and recording camera vibrations and movements. These movements can be used to align camera frames. Camera vibrations can be compensated by either translating subsequent frames by the measured camera movement and known geometry, or by aligning the image frames based on fixed features in the image but outside the subject's body.

Extracting Subject Position

In order to quantify measurements it is advantageous to know the subject position at any point in time. The available assigned positions may be for example, the following ones: Supine (belly-up), Prone (belly-down), On right side, On left side, Out of bed.

The set of subject positions may be enlarged to incorporate intermediate positions and sitting positions. Each position can be defined according to the visibility of each marker in that position. So for a particular subject position, there will be a subset of markers that must be seen (vis=1), a subset that cannot be seen (vis=0) and another subset that may be seen (vis=0.5). At any given time the subject position can be deduced by assigning the nearest subject position, where the distance to subject position i is given by: d _(i)=Σ_(j=1) ^(N) |vis _(j) ^((i)) −vis _(j)| where vis_(j) is the visibility of marker j in the current subject posture and vis_(j) ^((i)) is the position of that marker in subject position i. The reference values of the visibility in each subject position can be set once a priori provided the markers and camera are placed similarly from subject to subject. Note that the anatomical position of each marker is pre-set and determined according to some unique attributes of the marker such as: size, shape, wavelength emitted, temporal pattern, etc. Extracting Non-breathing Motion

The marker movements consist of respiratory movements, cardiac movements and other gross movements unrelated to the cardio-respiratory system. It is important for some applications such as sleep studies to identify periods of such movements, extract them, and continue measuring breathing parameters in their presence when possible. Such movements may include leg movements, twisting and turning in bed, bed-exit and others. The respiratory analysis may be carried out differently during periods of body movement and therefore these periods need to be identified.

As the markers are tracked one can identify frames in which there was significant non-breathing motion as follows: For each marker at each frame, the image positions of the marker in all the recent frames (say, 10 seconds) are analyzed through principal component analysis. When the body motion is pure breathing, the magnitude of the first principal component of each marker is significantly larger than the magnitude of the second principal component. The direction and absolute size of the first principal component vector is dependent on the marker location, relative camera position and subject's physiology. However, what is found to normal breathing motion is that locally it is along a fixed 3D axis throughout the entire cycle. By choosing a camera position and marker positions whose lines of sight are not parallel to these 3D physiological breathing directions, the local directionality translates to the 2D image. In the image coordinates, breathing motion is characterized by locally possessing a principal component that is significantly larger than the second one in the 2D image. Once the magnitude of the second principal component becomes significant relative to the first principal component for a few frames (for 0.5 seconds for example) of any chest/abdominal marker, it is an indication that some kind of significant non-breathing motion has occurred. At this stage tracking is reset and detection is carried out again. In cases where the camera line of sight is close to parallel with a marker breathing axis, the above Principal Component Analysis should be carried out in 3D (see below how the 3D vector is reconstructed).

An additional indicator of non-breathing motion is obtained by counting the number of pixels who have changed their gray level significantly (greater than a fixed threshold) from frame to frame of the video sequence. The number of pixels which obey this criterion during breathing can be determined from a baseline period, say, 30 seconds when the subject being monitored is not moving significantly (asides from breathing motion). Using the statistics during the baseline period, the threshold on the number of pixels that change significantly for identifying non-breathing motion is set.

Once the single marker criterion described above or the overall frame to frame difference is identified, an epoch of non-breathing motion is found. The epoch ends once the markers are redetected and can be tracked for a short period without non-breathing being detected.

2D to 3D Motion Tracking

The 3D displacements of the markers in the body frame of reference needs to be determined in order to calculate various parameters such as breathing volumes. With a 3D camera, the additional depth coordinate of each pixel is provided. However, these coordinates are in the camera frame and not relative to a body frame of reference. The following description illustrates how the 3D motion can be obtained using only the 2D image coordinates and a marker that consists of 2 or more LEDs.

One possible marker configuration includes 2 LEDs fixed to a rigid plate and separated from each other by a known distance, say 4 cm. The structure of the marker can be that of FIG. 6 with electronic circuitry and battery source that powers 2 LEDs. Each of the 2 LEDS are detected and tracked as described above. In order to calculate their 3D displacement and vector direction, consider two video frames in time: the end of an exhalation (beginning of inhalation) and the following end of inhalation. These frames can be found by calculating 1D curves of the marker displacement along the instantaneous principal component direction and subsequently analyzing these marker displacements curves for minima and maxima using standard 1D methods. The shape of the quadrilateral formed by the four 3D positions of the 2 LEDS (two at the beginning of inhalation and two at the end), is well-approximated by a parallelogram. This is due to the physiological characteristic that the breathing vectors are more or less fixed over small body surface regions (say 5×5 cm) of the abdomen or chest. One condition is not to locate the markers over points of discontinuity such as the ribcage-abdomen boundary. The base of the parallelogram has fixed length since it on a rigid marker but the length of its sides is unknown. First we describe how the length is determined assuming the parallelogram angle was known. For this task we use well-known camera pose estimation techniques, which are used to determine the 3D camera position of a rigid body given the image coordinates of at least 3 known points in the object coordinate system (see for example POSIT algorithm in Dementhon and Davis, International Journal of Computer Vision, Vol. 15, June 1995). The above described parallelogram can be regarded as a virtual “rigid body”—two of its vertices possess image coordinates obtained from the frame at the beginning of the inhale while the two other vertices are determined from the frame at the end of the inhale. In the object frame, the coordinates of the vertices relative to a vertex at the beginning of inhale are

-   -   1. (4,0,0)     -   2. (h cos θ, h sin θ, 0)     -   3. (4+h cos θ, h sin θ,0)         where θ is known, but h is yet to be determined. Had h been         known, the camera pose could be determined using standard         methods. So a set of values for h are assumed, say in the range         of 1-20 mm, and thus a set of corresponding camera poses are         determined. The correct h is determined as the one which leads         to minimal least square error between the measured image         coordinates from the tracking and the coordinates determined         from the corresponding camera pose. In a similar manner the         angle θ can be varied in addition to h in order to determine the         camera pose with least square error. Thus the angle can be         determined without prior knowledge.

Once the breathing vector from the beginning of inhale to the end of inhale is found, the displacements for intermediate frames can be found from the principal component displacement graphs by simple linear scaling. Specifically, for each monotonic segment of the 1D graph of FIG. 11, the following transformation is carried out from left to right: y′(t)=y′(t1)+factor*(y(t)−y(t1)) where factor=Y3d(t1,t2)/(y(t2)−y(t1)). The function y(t) is the 1D original principal component projection at frame t, and y′(t) is the 3D displacement graph at frame t. A monotonic segment starts at frame t1 and finishes at frame t2. y′(t=1) is set to y(t=1) and Y3D(t1,t2) is the length of the 3d vector corresponding to h found for the breathing maneuver from frame t1 to frame t2.

In order to increase the accuracy of determination of the 3D displacements, markers with more than 2 LEDs can be used as follows:

-   -   1. In a first frame, say the beginning of inhale, the 3D camera         position is determined using standard methods based on the         knowledge of 3 or more points on the rigid body (the marker).     -   2. The positions of the LEDs in a subsequent video frame (end of         inhale) are assumed to all be translated from their original         positions at the beginning of inhale by a fixed 3D vector         [a,b,c] that represents the breathing vector. Using the camera         pose determined from the first frame (Rotation and Translation)         along with the image positions of the LEDs in the second frame,         the values of a, b and c can be determined by least squares.

In order to limit the power consumption or the use of multi-LED markers throughout the monitoring, the following methods can be used:

-   -   1. Prior knowledge learned by using 3 or more LEDs on a marker         at specific body locations on a set of training subjects. For         example, it was found that the breathing vector for an abdominal         marker along the central superior inferior body axis is close to         being perpendicular to the marker's rigid plane.     -   2. Learning breathing angles on a training set of subjects using         multi-LED markers and applying the learned results to real-time         monitoring of subjects using 2-LED markers. These learned angles         are determined for a number of subject breathing patterns and a         number of marker placements on the training subjects. The         relevant vector directions according to body type (to be         described hereinafter), anatomical location, body positioning         and breathing type (deep or shallow for example) can be used         once a test subject is monitored. Alternatively, a         representative vector direction for a wide-range of conditions         (approximate surface region, all breath types, etc.) is often         sufficient.     -   3. Use 3 or more LEDs on each marker on subject in question         during baseline setup or intermittently. These can be the same         markers that are used during monitoring; in order to limit power         consumption the extra LEDs are turned off during regular         monitoring or flash in some time dependent way to reduce power         consumption. During the baseline setup all the marker LEDs are         used to determine the vector directions; subsequently only 2 or         more of them are used to determine the vector displacement.         -   Another configuration in which 3D motions can be             reconstructed is by using several distinct markers. For             example, several single marker LEDs can be distributed at 2             or more distinct locations on the chest and the distances             between them measured using measuring tape or some other             measurement means at a particular time in the respiratory             cycle (say end of exhale). From these measurements, the             camera position can be determined. Using the directions of             breathing motion at each of these locations (determined             using methods such as those described herein), the breathing             displacements can be estimated analogously to the way             described for the multi-led markers using image frames at             two different times (t1 & t2) in a single respiratory cycle,             say beginning of inhale and end of inhale. Here the “virtual             rigid body” that is being tracked is the 3D slab formed by             the vertices of the marker locations at t1 (define one face             of slab) and the marker locations at t2 (define second face             of slab). In the specific case where there are only 2 single             markers, the virtual slab reduces to a quadrilateral with             two long almost parallel sides of known length.             Sensor Data Fusion:

The video analysis and inertial sensor measurements of breathing motion can be combined to provide an improved estimate of the respiratory displacements. This can be achieved by utilizing the video measurements of displacements in the above described Kalman filter. In order to realize this combination, the relative rotation between the camera frame of reference (fixed) and that of the accelerometer should be considered. Furthermore, the markers and the accelerometer must be on the same rigid base or very close to each other, so that they actually measure the same body movement.

Missing Data:

Asides from achieving more accurate results by fusing measurements from both the imaging and the inertial sensor, the two sensors can be used to fill in measurements when one of the sensors cannot make a measurement, when there is no line of sight between the imaging sensor and the LEDs, or when there if a partial obstruction of some of the LEDs. This can be achieved by using the previously described Kalman filter with measurement covariances, which are dynamically changed according to the quality of the data measurements. For example, when a LED becomes completely obstructed, the covariance of its measurements is set to the maximal value for the filter iterations.

3D Surface Visualization

The extracted respiratory motion can be visualized in 3D as follows. First a 3D surface model representing the subject's body is extracted. If the camera is a 3D depth camera, the surface geometry is available from the depth measurements. In the case of a 2D camera, there are several options:

One possibility is to use two or more image sensors or to move the camera along a rail in order to obtain two or more views of the subject. If only a single camera is used the views will not be a true stereo pair. However, by considering breaths of the same 3D magnitude, similar duration and similar neighboring breaths but taken from two different camera positions, corresponding frames can be considered as stereo pairs, provided there was no significant non-breathing motion during the acquisition. Thus for example the beginning of inhale frames are considered a stereo pair and from there can progress through entire breath. The artificial stereo pairs can be viewed remotely using a 3D viewing device.

Another possibility is to register the stereo images to each other in order to get a 3D model of the person's trunk. For the registering one needs to solve the correspondence problem (see for example D. Scharstein and R. Szeliski. A Taxonomy and Evaluation of Dense Two-Frame Stereo Correspondence Algorithms, International Journal of Computer Vision 47(1/2/3) 2002), which can be typically either correlation or feature based. The LED markers could be used as features. In addition additional features can be obtained by using additional markers of other types, such as strips printed with a retro-reflective pattern, or clothing or a sheet with patterns visible by the image sensor. The sheet or clothing can be adjusted to fit snug around the patients trunk for the duration of the stereo acquisition. The requirements on these features is that they remain at relatively constant body locations only for a short period (say 1 minute) as opposed to long time monitoring where adhesion of marker to subject's body is needed.

Once the 3D surface model is extracted from a single stereo pair, the 3D respiratory motion extracted from each of the markers can be visualized by deforming the surface model by an amount which is scaled (by a factor greater than 1) compared to the extracted respiratory marker measurements. In order to show the evolution of the entire surface model, interpolation and extrapolation is performed between the markers and extended to the trunk boundary.

Once the breathing displacements at the marker positions are determined, an entire field of displacements can be obtained by interpolation and extrapolation to the subject's trunk. The subject's trunk position can be determined through segmenting a video frame. This can be carried out through standard region growing methods using the trunk markers as anchors and pixels with a significant optical flow vector as candidate region pixels. The optical flow is calculated between two or more frames representing large deviations of the respiration cycle (peak exhale to peak inhale for example). Also markers known to belong to other body parts such as those on a leg or arm should be excluded from the grown region.

The 3D surface of the extracted subject's trunk can be subsequently extracted using the methods described hereinabove. The dynamics of the trunk surface can be shown in real time providing a 3D visualization of the breathing movement. The color of each pixel is proportional to its breathing displacement at that time.

Such visualization can be streamed automatically to a remote device (nurse station or caregiver's mobile device), once an alert is triggered to the remote device. Conversely, a caregiver can use this view remotely any time they want to check on how the subject is doing. Several visualization schemes are possible in these situations, for example:

-   a. The normal camera view; -   b. The magnification of movement view described above; -   c. A different color display, dynamic range, different illumination     or different camera filter which are geared towards enhancing the     view of the subject rather than the markers attached to him; -   d. Control over viewing angle and position of the camera to observe     scenes surrounding the subject that were out of view.     Calibrations & Baselines:     Camera Calibration:

For accurate reconstruction of 3D marker and subject coordinates, it may be necessary to calibrate the camera once first connected using known grid pattern targets (without a subject). In some embodiments the camera is used at two positions to mimic stereo acquisition. Here too the camera should be calibrated with respect to the fixed known physical baseline (for instance, a rod placed on the bed). These calibrations can be achieved by many available tools (see for example: http://www.vision.caltech.edu/bouguetj/calib_doc/htmls/example5.html).

Camera Positioning:

The video sensor continuously tracks the markers that are positioned on the subject. The field of view should be set to include the markers from a perspective that most pronouncedly reveals the breathing movement. The determination of breathing displacements that are perpendicular to the image plane will usually be less accurate than those that lie in the image plane due to geometrical considerations. According to one embodiment of the invention, to optimize the camera position (or positions in the presence of more than a single camera), calibration is carried out. The camera positions and fields of view can be set-up manually or automatically. For automatic set-up, a calibration procedure is used. One such exemplary process for the case of a single camera is as follows:

-   -   1. The camera is positioned to include at least one marker in         its field of view and put into focus.     -   2. Once the focus is set, a predetermined set of viewing angles         is used to collect short video-clips of the subject. In each of         these video-clips the markers are detected and tracked.     -   3. The optimal camera viewing angle is now chosen. The main         criterion for this choice is related to which markers are         visible in the field of view. For each video clip i, the         following score is calculated:         Score_(i)=Σ_(m)W_(m)     -    where the sum is over all the visible markers m and W_(m) is a         predefined weight per marker. The identity of each visible         marker can be deduced from the video clip either according to         properties such as its shape, its relative position to other         markers or its color (using the camera's visible light sensor).         The identity of a marker can also be deduced from its         time-dependent signal. For example, each marker can have a         distinct predetermined frequency with which it flashes between a         minimal and maximal intensity level. The above calibration         procedure is carried out initially at set-up or whenever         tracking of the current markers has been aborted such as when         subject walks out of the room or when the bed and camera move         relative to each other. The subject need not sit still during         the set-up but obstructing the line of sight between the camera         and subject is to be avoided to shorten collection time.         Personalization and Volume Calibration:

With the above system, various movements can be tracked and analyzed. In order to convert these movements to physiological quantities, some calibration may be necessary. In one embodiment of the invention the model for breathing volume measurements is calibrated as described hereinafter, to take into account the characteristics of the subject and can be automatically recalibrated with changes in both subject position and with breathing patterns. The methods described up to now allow for the determination of 3D displacements of the chest wall but not of breathing volumes. The possible methods for automatic calibration of breathing volumes are described in the following.

Spirometer-based Calibration Per Subject:

The normal baseline tidal volume for an individual depends on many factors such as: age, sex, obesity, pose, disease conditions, time evolved since anesthesia, etc. One possibility for calibration of baseline tidal volume of a subject is to carry out a calibration procedure of the above video device versus a spirometer which measures flow rate of air inhaled and exhaled. In order to verify how the markers move at different tidal volumes and in different poses, additional calibration measurements are needed to be carried out on the subject. These tests can be done prior to surgery or to an invasive or imaging procedure, for patients who are to be monitored during or post procedure/surgery.

Limited Spirometer Based Training Set Followed by Personalization:

An alternative to the calibration per subject is to learn/build a model on several subjects or subject simulators a priori in a training session in which several breathing patterns and poses are recorded both with the video device and a spirometer. In addition, other physiological data is collected on these training subjects using the camera, specific covering (for example a snug sheet or undergarments only) and possibly additional markers. In particular the following information can be collected:

-   -   1. Silhouette of the trunk area which moves during breathing:         This is achieved in a number of ways but most simply by looking         at differences of video frames taken at opposite phase during a         respiratory cycle when the scene is illuminated. Pixels which         change significantly more than they do for consecutive frames         participate in respiration. These pixels can be pot processed to         form a smooth region by region growing techniques.     -   2. 3D shape of subjects trunk using either a large number of         markers as described above or using shape from shading         techniques with specific ambient illumination (B. Horn.         Obtaining shape from shading information. In P. Winston, editor,         The Psychology of Computer Vision. McGraw-Hill, New York, 1975).         Other methods of determining 3D shapes of human bodies can be         used (see for example: B. Allen, Learning body shape models from         real-world data, PhD Thesis, University of Washington, 2005).         These are based on using a 3D camera in the training set from         which the 3D body shape can be extracted. Furthermore the         dominant features that differentiate between body shapes can be         identified and mapped onto features that can be deduced from a         regular 2D camera. For a test subject, these 2D features can be         collected in a calibration session and mapped to find the         relevant 3D body shape.     -   3. Various distances between body landmark features and/or         markers and/or features of covering.

For each subject the spirometer volumes are fit to a function of the marker displacements and phases. The number of data points entering the fit is according to the video and spirometer sampling rate, which is of the order of tens of samples per second, so that say a 5 minute trial involves several thousand data points. This is repeated for additional subject poses (lying on the side, for example). The volume function can be the result of a non-linear regression scheme, e.g. SVM regression (B. Schölkopf and A. J. Smola. Learning with Kernels. MIT Press, 2002).

This model is kept in a database that can be accessed by the computer device of the breathing monitor. The model learned can be a linear function of the displacements; the phases can be taken into account by using signed displacements which are determined by setting a reference zero displacement for each marker. One way the zero is assigned is by computing a moving average of the raw displacements over the last few respiratory cycles (say 5 breaths for example). The signed displacement is obtained by subtraction of the reference zero displacement from the measured one at each time point. The relative weights between markers can be based on their relative “reliability” which is provided by the variance in equi-volume measurements or the variance of displacements during stable tidal breathing.

Alternatively, the phases can be taken into account by learning different models for different scenarios of phase relations between the markers. For example one function for all markers in-phase, another one for chest and abdominal markers out of phase, and so forth. Once a new subject is to be monitored, a short automatic calibration session (e.g., 30 seconds) is performed according to a predefined protocol such as “subject lies on back” is undertaken at which time various relevant measurements are made for an estimate of the subject's silhouette and 3D shape as in the training set. In addition other features related to trunk size are measured: These could be the distances between markers for example which can be extracted using the device's image frames or distances between body landmarks identified through image analysis or manually. In order to convert these distances to actual length units, e.g., cm, on or more objects of known dimensions are included in the field of view or on the body for calibration purposes.

Using the data gathered on the current test subject, the “nearest-neighbor” database subjects are extracted from the database. The distance between two subjects is measured for instance by measuring the distance between their trunk images (within silhouette) as follow:

-   -   1. The two images are registered using body landmarks and         markers locations placed in corresponding anatomical positions.         The registration is non-rigid but can be confined, for instance,         to Affine transformations.     -   2. The distance between 2 images is determined by their scale         factors (for example by the difference of the geometric mean of         the scale factors to 1).

Once the relevant data base subjects that match the current subject the closest are found, the dependence of spirometer volume on 3D marker displacements found for the database subject is utilized to estimate volume. The 3D vector displacements are scaled by the scale factors before being entered into the function associated with the database subject.

Change of Position: The marker movements and breathing volumes change as a function of subject pose. The pose of a subject is deduced in real time as described herein. For each subject position, the relevant database estimation for the volume is used. However, the relevant “nearest neighbors” from the database are selected during the initial auto-calibration session of the test subject.

FIG. 1 is a flow chart of the monitoring step that starts with the inputs of the first few frames 101 after which, in step 102, the markers are detected according to the procedure of FIG. 2. The detection step is followed by the impact of the next frame 103 after which, step 104 the markers are tracked over time as detailed in FIG. 3. From the frames the subject's pose is extracted in step 105, and the respiration properties are calculated in step 106, as further detail in FIG. 4. At the end of the calculation process the system checks, 107, if the tracking has been lost and in the affirmative case the process is started anew in step 101, while in the negative case the next frame is input in step 103.

FIG. 2 details the detection steps. At the beginning of this step the system already has at least one frame and the detection starts by getting the new frame, 201. A comparison step 202 is performed to determine whether the frame is stable compared to the previous one. In the negative case another frame is obtained and the process starts again at step 201, while in the affirmative case the frame is set as a reference frame and in step 203 all pixels greater than a threshold value are located. The pixels are then clustered into “clouds” in step 204 and the clouds are filtered according to their features in step 205. In step 206 the markers (clouds) are tracked from the reference frame to the current frame and the success of this tracking is verified in step 207. If the tracking was unsuccessful the process starts again at step 201, while if it was successful after t seconds have elapsed since the reference frame (verified in step 208) tracking of markers whose motion is consistent (similar frequency) is continued in step 209, and the analysis of marker movement begins in step 210. If the time that passed is less than t seconds, the next frame is obtained in step 211 and the markers are checked again in step 206.

FIG. 3 is a flowchart of the tracking procedure, which starts by setting the first frame is a reference frame at 301. Then, the next frame is obtained, 302 after which the system checks, in step 303, whether t seconds have passed since the reference frame, in which case the last frame is set to be the reference frame in step 304. In the negative case, step 305 checks whether the brightness has changed relative to the reference frame, in which case, again, the frame is set to be the reference frame in step 304. In the negative case, the 2-D motion vectors are calculated between the current and the reference frame in step 306. This also happens after the last frame has been set to be the reference frame in step 304. The system then checks whether the tracking is successful using criteria such as making sure the extent of the marker movement is not too large, that the pixel movements within the marker are consistent, and that the mean square error in the numerical fit is not too large in step 307, and in the affirmative case it goes back to step 302 and gets the next frame. If tracking was unsuccessful the detection process of FIG. 2 is performed again in step 308.

FIG. 4 is a flowchart of the analysis process. In this process would begin by getting 2D tracking locations of markers, in step 401. Step 402 calculates the projection of marker location onto the current principal component direction from the last t seconds. If a new peak is identified in the principal component function over time, in step 403, the 3D position of the marker at the time of “New Peak” frame is estimated in step 404; else, step 401 is repeated. The 3D position of the marker at all frames between the two most recent peaks is estimated by scaling the principal component graph in step 405 and the RR, phase delay and the volumes for all frames between the most recent two peaks is calculated in step 406, after which the process restarts at step 401.

FIG. 5 is a flowchart of a volume calibration procedure that may be carried out according to some embodiments of the invention. The calibration process begins in step 501 where mechanical calibration is performed. Then, in step 502, the subject is recorded from each camera in supine position in various illumination conditions. In step 503 the marker's position on the subject are tracked to obtain 3-D displacement vectors, and in step 504 the silhouette and the 3D shape of the subject's trunk is extracted. In step 505 the “nearest neighbor” database subject is found by registering the silhouette and the 3D shape via a non-rigid transformation. In step 506 the scaling factor between the subject and his nearest neighbor is determined. The calibration function is found in step 507 by using the nearest neighbor estimator function with the current subject's scaled displacement.

Illustrative Markers

FIG. 6 illustrates the main parts of a marker according to one embodiment of the invention, consisting of LED assembly 601, which contains a LED, together with circuitry needed to operate it. Operating LED's is conventional in the art and is well known to the skilled person and therefore said circuitry is not shown in detail, for the sake of brevity. Assembly 601, in this embodiment of the invention, is meant for multiple uses and operates together with disposable part 602, which in some embodiments of the invention is applied directly to the subject's skin, e.g. using an adhesive surface provided at its bottom. In the embodiment of FIG. 6, however, disposable part 602 is located above a fabric 603, which may be, for instance, a subject's pajama or a blanket, and is kept in position relative to said fabric by a lower base 604, which is maintained in position on the subject's body by using an adhesive material located on its bottom 605. Lower base 604 can be coupled to the bottom 606 of disposable part 602 in a variety of ways, e.g. by magnetic coupling between the two, or by providing mechanical coupling, e.g., through pins that perforate and pass through fabric 603 (not shown).

As will be described in greater detail with reference to FIG. 10, disposable part 602 may contain batteries needed to operate the LED, as well as, if desired, additional circuitry. Reusable LED assembly 601 is connected to disposable part 602 through seats 607 and 607′, which engage buttons 608 and 608′ located in disposable part 602. An additional positioning pin 609 can be provided in LED assembly 601, to engage hole 610 in disposable part 602. Electrical contact can be provided from the power supply located in disposable part 602, through buttons 608 and 608′, although of course many alternative ways exist of conveying power from the batteries to the LED assembly 601.

FIG. 7 is an operationally connected view of the various elements of FIG. 6, the same numerals being used to indicate the same parts. FIG. 8 is a top view of the assembled marker devise of FIG. 7, and FIG. 9 is a cross-section of the same device assembly, taken along the AA line of FIG. 8. In this cross-section, in addition to the elements already described above a compartment 611 is seen, which houses the LED, as well as electronics used to operate it. Of course, the top portion 612 of LED assembly 601 is made of material of a transparency sufficient to allow the required amount of light generated by the LED to be viewed from the outside.

FIG. 10 shows the disposable part 602 of FIGS. 6-9 in exploded view. This disposable assembly is made of a plurality of elements and layers, kept together by gluing or by mechanical connection, for instance when upper buttons 608 and 608′ are connected with a lower buttons 613 and 613′. A battery 614 is provided as the power supply to the LED assembly, which can be, for instance, a standard 2030 disc battery.

As will be apparent to the skilled person the arrangement of the disposable assembly 602, as well as that of the LED assembly 601 provided in the figures, represents only one of countless possible arrangements and is provided only for the purpose of illustration. Additionally, Multi-LED assemblies can be devised by the skilled person, to provide two or more LEDs in the same LED assembly, while performing the required dimensional changes. Such alternatives and modifications are clear to the skilled person from the above description and, therefore, are not further described in detail, for the sake of brevity.

All the above description of markers, methods, uses, etc. have been provided for the purpose of illustration and are not intended to limit the invention in any way. Many modifications, alternative configurations, process steps and methods can be provided by the skilled person, without exceeding the scope of the invention. 

The invention claimed is:
 1. A system for monitoring the respiratory activity of a subject, comprising: a) one or more signal generating elements being kinematic or light emitting elements, applied to the thorax of a subject, for generating signals that are indicative of movement of the thorax of said subject; b) a camera for receiving said generated signals during breathing motion of said subject; and c) one or more computing devices in data communication with said receiver, for analyzing said breathing motion, wherein said one or more computing devices is operable to: i. calculate, in response to said received generated signals, a magnitude of a maximum displacement in 3D space of said one or more signal generating elements during a cycle of said breathing motion, wherein the maximum displacement is related to first and second reference breathing vectors associated with a same respiration cycle of said subject, said second reference breathing vector being complementary to said first reference breathing vector and originating from a maximal light emitting element location corresponding to exhalation conditions and terminating at a minimal light emitting element location corresponding to inhalation conditions; calculate a reference tidal volume that is related to said first and second reference breathing vectors; iii. store data associated with said reference tidal volume in memory; iv. calculate in a monitoring stage, for each of said one or more light emitting elements, the magnitude of a current displacement in 3D space of said one or more light emitting elements during said breathing motion related to corresponding complementary first and second monitored breathing vectors that are indicative of thorax movement and extend between a minimal light emitting element location and a maximal light emitting element location associated with a same monitored respiration cycle of said subject; v. disregard one or more of said second monitored breathing vectors if a motion deviation of recent sampled light from a single direction exceeds a predetermined threshold; and vi. calculate a value of a current tidal volume for said monitored respiration cycle by determining a function of the magnitude of reference displacements and reference tidal volumes, when the corresponding second monitored breathing vector is non-disregarded, with respect to the magnitude of a corresponding maximum displacement, and multiplying said function by the value of the corresponding reference tidal volume.
 2. The system of claim 1, wherein each of the one or more kinematic signal generating elements is an accelerometer, a gyroscope, or a combination thereof.
 3. The system of claim 1, wherein each of the one or more signal generating elements is a light source assembly which comprises two or more spaced light sources suitable to generate light, which are mounted on a rigid plate and fed by a power source, and the receiver is a stationary camera for tracking movement of, and imaging, said two or more light sources during said breathing motion of the subject.
 4. The system of claim 3, wherein the light source assembly is adapted to generate light signals of a pre-assigned time dependent intensity.
 5. The system of claim 1, wherein: each of the signal generating elements is disposable and holds power supply elements and operating circuitry; each of the signal generating elements is provided with an adhesive and is suitable to be positioned directly on the subject's thorax; each of the signal generating elements is positionable on a fabric located in close proximity to the subject's thorax, and is further provided with fastening means adapted to maintain it in a desired positional relationship to a chosen location on the subject's thorax; or each of the signal generating elements is connectable to a base attached directly to the subject's thorax, wherein a fabric is located between said base and the signal generating element.
 6. The system of claim 4, wherein the pre-assigned time dependent intensity is a known frequency and amplitude, is selected from a frequency or amplitude, or is of time-dependent intensity different from that of at least another one of the light sources in the system.
 7. The system of claim 1, which is provided with connection lines suitable to deliver raw or analyzed data to a remote device.
 8. The system of claim 1, wherein the signal generating elements include a first group of one or more kinematic signal generating elements selected from an accelerometer, a gyroscope, or a combination thereof, and a second group of signal generating elements including one or more light sources or one or more light source assemblies, each of which comprises one two or more spaced light sources suitable to generate light, which are mounted on a rigid plate and fed by a power source, and the receiver is a stationary camera for tracking movement of, and imaging, said two or more light sources during breathing motion of the subject, the one or more computing devices operable to calculate the magnitude of a maximum displacement and of a current displacement in conjunction with both said first and second group of signal generating elements.
 9. The system of claim 1, wherein the one or more computing devices is also operable to output an alert when the current tidal volume for a monitored respiration cycle deviates from the reference tidal volume by more than a second predetermined threshold that is indicative of an anomalous breathing pattern.
 10. A method of calculating a subject's breathing volume, comprising the steps of: a) providing on the thorax of a subject a marker comprising one or more signal generating elements fixed to said base, for generating signals that are indicative of movement of the thorax of said subject; b) by a receiver, receiving said generated signals from said one or more signal generating elements during breathing motion of said subject; and c) analyzing said received generated signals by one or more computing devices in data communication with said receiver, said analysis comprising the steps of: i. calculating, in response to said received generated signals, the magnitude of a maximum displacement in 3D space of said one or more signal generating elements during a cycle of said breathing motion; and ii. calculating the magnitude of a current displacement in 3D space of said one or more signal generating elements during said breathing motion with respect to a reference tidal volume associated with said maximum displacement in 3D space, wherein each of the one or more signal generating elements is a light source assembly which comprises two or more spaced light sources suitable to generate light, and the receiver is a stationary camera for tracking movement of, and imaging, said two or more light sources during breathing motion of the subject, the method further comprising the steps carried out by the one or more computing devices of: iii. calculating, in response to receiving imaged data taken by said camera at a predetermined orientation, the magnitude of a maximum displacement in 3D space of said two or more light sources during said breathing motion related to first and second reference breathing vectors associated with a same respiration cycle of said subiect, said second reference breathing vector being complementary to said first reference breathing vector and originating from a maximal light source location corresponding to exhalation conditions and terminating at a minimal light source location corresponding to inhalation conditions; iv. calculating a reference tidal volume that is related to said first and second reference breathing vectors; v. storing data associated with said reference tidal volume in memory; vi. calculating in a monitoring stage, for each of said two or more light sources, the magnitude of a current displacement in 3D space of said two or more light sources during said breathing motion related to corresponding complementary first and second monitored breathing vectors that are indicative of thorax movement and extend between a minimal light source location and a maximal light source location associated with a same monitored respiration cycle of said subject; vii. calculating in a monitoring stage, for each of said two or more light sources, the magnitude of a current displacement in 3D space of said two or more light sources during said breathing motion related to corresponding complementary first and second monitored breathing vectors that are indicative of thorax movement and extend between a minimal light source location and a maximal light source location associated with a same monitored respiration cycle of said subject; and viii. calculating the value of a current tidal volume for said monitored respiration cycle by determining a function of the magnitude of reference displacements and reference tidal volumes, when the corresponding second monitored breathing vector is non-disregarded, with respect to the magnitude of a corresponding maximum displacement, and multiplying said function by the value of the corresponding reference tidal volume.
 11. The method of claim 10, wherein: the reference tidal volume is obtained using a spirometric measurement; the reference tidal volume is a volume related to a subject in a training set, or a volume related to two or more subjects in a training set; or the reference tidal volume is stored in a database and is representative of a plurality of previously acquired breathing patterns of subjects that have similar features to those of the monitored subject.
 12. The method of claim 10, wherein a number of data points used for fitting the corresponding training breathing vector is according to a video or spirometer sampling rate.
 13. The method of claim 10, where the training sets differ from each other by amplitude and by phase.
 14. The method of claim 10, further comprising the step of outputting an alert when the current tidal volume for a monitored respiration cycle deviates from the reference tidal volume by more than a second predetermined threshold that is indicative of an anomalous breathing pattern, for personalizing an alarm related to respiratory activity of a subject.
 15. The method of claim 10, wherein the alarm is outputted only whenever base movements are uni-directional.
 16. A method according to claim 10, further comprising: a) providing on the abdomen or on a different location of the thorax of a subject, an additional marker comprising at least two spaced light sources which are mounted on a rigid plate and fed by a power source; b) generating light from said at least two light sources of said additional marker; c) tracking movement of, and imaging, said two or more light sources of said additional by the camera, during breathing motion of said subject; and d) analyzing said tracked movement by the computing devices, said analysis comprising the steps of: i. calculating, in response to receiving imaged data taken by said camera and during said breathing motion, the phase difference between corresponding displacements in 3D space of the light sources on said sources marker and the light sources on said additional marker; ii. calculating the breathing volume based on said displacements in 3D space and said phase difference; and iii. obtaining indications regarding obstructive breathing of said subject if said phase difference exceeds a predetermined value.
 17. The method of claim 10, wherein the reference tidal volume is stored in a database and is representative of a plurality of previously acquired breathing patterns of subjects that have similar features to those of the monitored subject.
 18. A method according to claim 10, further comprising: a) providing on the abdomen or on a different location of the thorax of a subject, an additional marker comprising one or more kinematic signal generating elements; b) generating signals that are indicative of movement of the thorax of said subject; c) tracking movement of said one or more kinematic signal generating elements, during breathing motion of said subject; and d) analyzing said tracked movement by the computing devices, said analysis comprising the steps of: iv. calculating, in response to receiving said signals and during said breathing motion, the phase difference between corresponding displacements in 3D space of said one or more kinematic signal generating elements; v. calculating the breathing volume based on said displacements in 3D space and said phase difference; and vi. obtaining indications regarding obstructive breathing of said subject if said phase difference exceeds a predetermined value. 